Seismic Network Operations
Alum Creek State Park, Ohio, USA
US ACSO commences operations on: 2000,135
|Telemetry Status at the NEIC:||Last Data In Less Than 10 Minutes|
Site Geology: The Ohio Shale unit is a bituminous black shale with large concretions and is Devonian in age.
|Location Code||Channel Code||Instrument||Flags||Sample Rate||Dip||Azimuth||Depth|
PDF, Last Month
PDF, Current Week
Availability, Since 1972
Availability, 2 Month
As part of the annual calibration process, the USGS runs a sequence that includes a random, a step, and several sine wave calibrations. The USGS analyzes the random binary calibration signal in order to estimate the instrument response. The figures below show the results from the analysis of the most recent processed calibration at the station.
We use an iterative three-step method to estimate instrument response parameters (poles, zeros, sensitivity and gain) and their associated errors using random calibration signals. First, we solve a coarse non-linear inverse problem using a least squares grid search to yield a first approximation to the solution. This approach reduces the likelihood of poorly estimated parameters (a local-minimum solution) caused by noise in the calibration records and enhances algorithm convergence. Second, we iteratively solve a non-linear parameter estimation problem to obtain the least squares best-fit Laplace pole/zero/gain model. Third, by applying the central limit theorem we estimate the errors in this pole/zero model by solving the inverse problem at each frequency in a 2/3rds-octave band centered at each best-fit pole/zero frequency. This procedure yields error estimates of the 99% confidence interval.
|Loc||Chan||Cal Date||Epoch-Span||Grade||Amp Nominal Error (dB)||Amp Best Fit Error (dB)||Phase Nominal Error (degree)||Phase Best Fit Error (degree)||Sensor||Cal Type|
|00||BHZ||2014:071||2012:026 to No Ending Time||A||0.014541||0.014181||0.078965||0.071886||STS2-I||Random|
2012-08-18The seismic vault was repaired and instrumentation reinstalled.
2012-01-26Instrumentation moved inside building due to vault flooding.
2005-06-06Upgraded to Q330 digitizer.