Seismic Network Operations
IU BILL commences operations on: 1995,228
|Host:||Geophysical Survey, Russian Academy of Sciences|
|Telemetry Status at the NEIC:||Last Data In Less Than 10 Minutes|
Vault Condition: The vault lies below an apartment complex. The concrete pier is isolated from the floor.
Site Geology: Bilibino lies within the Anju fold region of the Chukotsk fold system. The Anjo fold region consists of ingneous rock such as gabbors-diabase and sedimentary rock including aleurite, argillite, crystal shale and various types of limestones.
|Location Code||Channel Code||Instrument||Flags||Sample Rate||Dip||Azimuth||Depth|
|00||VMZ||Streckeisen STS-1VBB w/E300||CH||0.10||0.00||0.00||0.00|
|00||VM2||Streckeisen STS-1VBB w/E300||CH||0.10||0.00||0.00||0.00|
|00||VM1||Streckeisen STS-1VBB w/E300||CH||0.10||0.00||0.00||0.00|
|00||LHZ||Streckeisen STS-1/VBB Seismometer||CG||1.00||90.00||0.00||0.00|
|00||LH2||Streckeisen STS-1/VBB Seismometer||CG||1.00||0.00||270.00||0.00|
|00||LH1||Streckeisen STS-1/VBB Seismometer||CG||1.00||0.00||180.00||0.00|
|00||BHZ||Streckeisen STS-1/VBB Seismometer||CG||20.00||90.00||0.00||0.00|
|00||BH2||Streckeisen STS-1/VBB Seismometer||CG||20.00||0.00||270.00||0.00|
|00||BH1||Streckeisen STS-1/VBB Seismometer||CG||20.00||0.00||180.00||0.00|
PDF, Last Month
PDF, Current Week
Availability, Last 30 Days
Current IssuesData flow was stopped from Sept. 2011 to mid-April 2012 due to non-technical reasons. Data flow resumed in April but stopped again in early July due to a broken network link. Data flow has resumed but data is noisy.
Current IssuesVertical STS-1 is serverly overdamped. Currently we do not have an agreement which alows us station access to fix the problem.
2014-11-16Upgraded to Q330 digitizer